â¢ In some regions apply diatomaceous earth insulation and the super thermal insulation material at low temperature, in a hazardous area area and base of the combustion chamber all for to cement the connection as much as possible using the 42% dense alumina refractory brick lining in the 6-hour high temperature during the test will provide sufficient insulation.
â¢ Fixation of available customized vertical test samples for testing
â¢ Refractory insulation brick used in heat-resistant lining, refractory filling block and the slag plate
â¢ Top shop special-shaped brick and lined with refractory material
â¢ Furnace using low carbon steel, reinforced with c channel steel and h-beams, steel resolve hot deformation
â¢ Sample fixed boxes mounted on the furnace front, fixed by four sets of doors
â¢ Cold noodles with inorganic fiber pad, heat-resistant insulated bricks along the wall pad, shop in situ high temperature block
â¢ Four observation port is located on the long side of the wall, use a heat-resistant quartz glass, clever, convenient operators watch the whole sample in a test case
â¢ Air cooler insulating frame lightweight aluminum fiberboard sliding doors without using a observation port, slide the door blocking the oven available in the heat
â¢ Twelve specialized luminous flame burner. Each fuel burners are fitted with flame supervision devices to ensure combustion combustion safety
â¢ Use button switches at any time automatically inserted into the mouth, according to the preset heat curve light burning mouth, such as the BS 476-DI20-24bu, EN 1363 and IMO the hydrocarbon curve. Manual control can be lit to specify burning mouth burning mouth.
â¢ The three genuine leaks smoke cross section is circular, each of which contains an airlock. These three air brakes is a round of high-level stainless steel structure and its shaft water cooling. Flue liner here for biodegradable ceramic fiber material.
â¢ Furnace using low carbon steel, reinforced with c channel steel and h-beams, solve the thermal deformation of the steel issue, designed as a strong deformation structures. Strengthening plates for burners, hole, thermocouples, pressure and monitoring points provide holes. Steel frame is designed to ensure that you have considered the use function. The durable nature of the equipment to ensure its life expectancy in a test environment.
In line with the latest regulations of the thermocouple, location and design.
In line with the latest regulations of pressure points, location and design.
LP gas industrial high temperature light soft flame burner.
Lifting frame and the fixed sample box.
Refractory lining of the connection pipe and chimney.
Burner air intake fan.
Temperature sensors and pressure gauge.
Motorized Pressure Controlled Dampers
The three air brake is driven by the motor,control the pressure of combustion chamber . The air brake has to be cooled by water and it requires customers to provide water and drainage.
PLC and computer control
Fire data management software is written according to customer's requirements to collect and store test data.It meets BS 476 part 20-24, EN 1363 and IMO hydrocarbon curve heating requirements.Other standard time temperature curves can also be programmed into the system in advance.The PLC controller includes a built-in operating interface, including the required recording and programming capabilities, as well as the necessary motor starter for all systems. The system programming can heat the curve in the real time and shows the boiler control system on computer screen.
Data recording system with dual recording
The device can spread its screen display on a standard computer network and any computer connected to the network can log in to the test information.The facility can be used to provide test data to observers in the laboratory.All records are automatically saved to another computer.
The furnace can be operated manually.The complete ignition system is not in any way dependent on the operation of the control computer.Press the clear start button to light the stove.A series of light instructions to clear the sequence correctly.Once they are on a separate burner Control Panel lighting up all the available at the pre-set time burner.If the furnace is not lit during that period, the system will have to be removed.Each region of the position of the air and gas damper motor can be adjusted individually or by a group of the main control regulation.During the test, the safety of the furnace is monitored by a set of interlocking devices, each of which has its own indicator light.The interlocking devices are closed, so that the linkage of the furnace closure is captured and maintained to the manual reset ignition sequence and an audible alarm device installed.